Opened 12 years ago
Closed 12 years ago
#26 closed enhancement (fixed)
Add "B[10.nV]" to Table 20
Reported by: | PaulSchluter | Owned by: | Gunther Schadow |
---|---|---|---|
Priority: | major | Milestone: | Revision 1.8 |
Component: | Keywords: | decibel bel logarithm | |
Cc: | gschadow@… |
Description
At a minimum, please add "B[10.nV]" to Table 20 so that decibel measurements can be expressed relative to this particular reference value [10.nV]. This also suggests a general approach for representing logarithmic Bel units "B[...]" relative to a reference value "...", where "..." is a valid UCUM string. It can be used to express other "dB" measurements relative to other scale factors and units-of-measure, such as electric field strength, in a consistent, composable and computable manner. This request was motivated by the need for the unambiguous expression of the units-of-measure for computed EEG parameters in the IHE PCD "Rosetta Terminology Mapping" (RTM) project.
Note that the "definitional unit" for Bel(P/Po) = log(P/Po) would be log(Po) for a reference power Po and that the "definitional unit" for Bel(A2/Ao2) = log(A2/Ao2) = 2log(A/Ao) for a reference amplitude Ao.
Change History (3)
comment:1 Changed 12 years ago by
comment:2 Changed 12 years ago by
Milestone: | → Revision 1.8 |
---|---|
Owner: | changed from gschadow@… to Gunther Schadow |
Status: | new → assigned |
comment:3 Changed 12 years ago by
Resolution: | → fixed |
---|---|
Status: | assigned → closed |
More explanation: The Bel (or decibel) represents a ratio of power levels (or power densities or other quantities clearly connected with power). Power levels are sometimes expressed as a voltage, P=U2/R. This is where the factor of two comes in when converting between decibel and the linear "definitional unit".
When the reference level is fixed, e.g. to 1 mW, the corresponding unit dBm represents an absolute quantity, rather than a dimensionless ratio. In the general case, the reference quantity is written in brackets behind dB: dB(20 μPa) or dB(mW/m2). Note that the conversion from logarithmic to linear involves a factor of 2 for "field units" like mV. The suggested formalism for adding required new UCUM codes for such a unit is to use the form B[...] where the reference quantity is included in square brackets as a valid UCUM code.